Contacts and Location
|Enquire about this equipment|
Riga Technical University
Faculty of Mechanical Engineering, Transport and Aeronautics
Institute of Biomedical Engineering and Nanotechnologies - Latvia, Rīga
Helios Spectrophotometers’ superior optical system provides stability, precision and linearity. All units feature angled casing, sealed keypad to protect from spills, optional printer, plug-and-play self-aligning lamps, large LCD, VGA graphics, full-
Wavelenght Calibration Method and Values: Filter Value 279.3; 360.8; 453.6; 573.0; 807.1 nm. Measured Value 278.2; 360.8; 453.0; 574.0; 807.0 nm. Absorbance accuracy Methods and Values: Filter Value 0.1030; 0.4493; 0.8884; 1.9504; 2.4356; 2.8726. Measured Value 0.1022; 0.4468; 0.8861; 1.9495; 2.4313; 2.8694
Thermo Spectronic (manufacturer's website)
Heλios Gamma, Delta
Training is required to use this equipment and we can arrange this if needed.
Yes, this equipment is calibrated.
Last Calibration: 16th August, 2006